The paper deals with continuous monitoring of NH3 desorption from material used as bedding in animal breeding establishments, using five semiconductor sensors SP-53 (FIS Inc.). In lab experiments, two groups of sensors from two categories, to which the manufacturer divided the sensors based on the value of their resistance RS, were used. Also the influence of air humidity over the values of NH3 concentrations measured by individual sensors was monitored. The study ascertained statistically relevant differences between NH3 concentration values measured by both groups of sensors, in the range from 8.5% to 23.1%. The biggest differences were confirmed for relative air humidity around 85%. It was verified that setting a convenient value of the load resistance RL with regard to the RS of each sensor, along with the selection of limited range of measured temperatures and relative air humidity, leads to satisfactory correction of the sensor output data, regardless of the division of sensors in categories pursuant to the value of their resistance in manufacturing.
load resistence; sensor resistence; air humidity; continuous NH3 monitoring
Barsan N., Weimar U. (2003): Understanding the fundamental principles of metal oxide based gas sensors: the example of CO sensing with SnO2 sensors in the presence of humidity. Journal of Physics: Condensed Matter, 15: 813–839.
Carotta Maria Cristina, Benetti Michele, Ferrari Elena, Giberti Alessio, Malagù Cesare, Nagliati Marco, Vendemiati Beatrice, Martinelli Giuliano (2007): Basic interpretation of thick film gas sensors for atmospheric application. Sensors and Actuators B: Chemical, 126, 672-677 https://doi.org/10.1016/j.snb.2007.04.016
Hahn S.H, Bârsan N, Weimar U, Ejakov S.G, Visser J.H, Soltis R.E (2003): CO sensing with SnO2 thick film sensors: role of oxygen and water vapour. Thin Solid Films, 436, 17-24 https://doi.org/10.1016/S0040-6090(03)00520-0
Hawkins D. (2005): Biomeasurement: Understanding, Analysing, and Communicating Data in the Biosciences. New York, Oxford University Press.
Henrich V.E., Cox P.A. (1996): The Surface Science of Metal Oxides. Cambridge, University Press.
Ionescu R, Vancu A, Tomescu A (2000): Time-dependent humidity calibration for drift corrections in electronic noses equipped with SnO2 gas sensors. Sensors and Actuators B: Chemical, 69, 283-286 https://doi.org/10.1016/S0925-4005(00)00508-6
Kamionka M., Breuil P., Pijolat C. (2006): Calibration of a multivariate gas sensing device for atmospheric pollution measurement. Sensors and Actuators B: Chemical, 118, 323-327 https://doi.org/10.1016/j.snb.2006.04.058
Korotcenkov G., Cho B.K. (2009): Thin film SnO2-based gas sensors: Film thickness influence. Sensors and Actuators B: Chemical, 142, 321-330 https://doi.org/10.1016/j.snb.2009.08.006
Masson N., Piedrahita R., Hannigan M. (2015): Approach for quantification of metal oxide type semiconductor gas sensors used for ambient air quality monitoring. Sensors and Actuators B: Chemical, 208, 339-345 https://doi.org/10.1016/j.snb.2014.11.032
Nakata S., Neya K., Takemura K.K. (2001): Non-linear dynamic responses of a semiconductor gas sensor – Competition effect on the sensor responses to gaseous mixtures.
Thin Solid Films, 391: 293–298.
Pavelko R.G., Vasiliev A.A., Llobet E., Sevastyanov V.G., Kuznetsov N.T. (2012): Selectivity problem of SnO2 based materials in the presence of water vapors. Sensors and Actuators B: Chemical, 170, 51-59 https://doi.org/10.1016/j.snb.2011.01.013
Tomic Oliver, Eklöv Tomas, Kvaal Knut, Haugen John-Erik (2004): Recalibration of a gas-sensor array system related to sensor replacement. Analytica Chimica Acta, 512, 199-206 https://doi.org/10.1016/j.aca.2004.03.001
Wang C., Yin L., Zhang L., Xiang D., Gao R. (2010): Metal oxide gas sensors: sensitivity and influencing factors, Sensors, 10: 2088–2106.