Evaluations of the effects of soil properties and electrical conductivity on the water content reflectometer calibration for landfill cover soils
K. Kim, J. Sim, T.-H. Kimhttps://doi.org/10.17221/158/2015-SWRCitation:Kim K., Sim J., Kim T.-. (2017): Evaluations of the effects of soil properties and electrical conductivity on the water content reflectometer calibration for landfill cover soils. Soil & Water Res., 12: 10-17.
This study presents soil-moisture calibrations using low-frequency (15–40 MHz) time domain reflectometry (TDR) probe, referred to as water content reflectometer (WCR), for measuring the volumetric water content of landfill cover soils, developing calibrations for 28 different soils, and evaluating how WCR calibrations are affected by soil properties and electrical conductivity. A 150-mm-diameter PVC cell was used for the initial WCR calibration. Linear and polynomial calibrations were developed for each soil. Although the correlation coefficients (R2) for the polynomial calibration are slightly higher, the linear calibrations are accurate and pragmatic to use. The effects of soil electrical conductivity and index properties were investigated using the slopes of linear WCR calibrations. Soils with higher electrical conductivity had lower calibration slopes due to greater attenuation of the signal during transmission in the soil. Soils with higher electrical conductivity tended to have higher clay content, organic matter, liquid limit, and plasticity index. The effects of temperature and dry unit weight on WCR calibrations were assessed in clayey and silty soils. The sensor period was found to increase with the temperature and density increase, with greater sensitivity in fine-textured plastic soils. For typical variations in temperature, errors in volumetric water content on the order of 0.04 can be expected for wet soils and 0.01 for drier soils if temperature corrections are not applied. Errors on the order of 0.03 (clays) and 0.01 (silts) can be expected for typical variations in dry unit weight (± 2 kN/m3).Keywords:
soil electrical conductivity; soil index properties; time domain reflectometry; volumetric water content; WCR calibrationReferences:
Anderson Stephen H., Udawatta Ranjith P., Seobi Tshepiso, Garrett Harold E. (2009): Soil water content and infiltration in agroforestry buffer strips. Agroforestry Systems, 75, 5-16 https://doi.org/10.1007/s10457-008-9128-3Arsoy Sami, Ozgur Mehmet, Keskin Erdinc, Yilmaz Cuneyt (2013): Enhancing TDR based water content measurements by ANN in sandy soils. Geoderma, 195-196, 133-144 https://doi.org/10.1016/j.geoderma.2012.11.019Campbell Gaylon S, Anderson Russell Y (1998): Evaluation of simple transmission line oscillators for soil moisture measurement. Computers and Electronics in Agriculture, 20, 31-44 https://doi.org/10.1016/S0168-1699(98)00006-4Chandler D. G., Seyfried M., Murdock M., McNamara J. P. (2004): Field Calibration of Water Content Reflectometers. Soil Science Society of America Journal, 68, 1501- https://doi.org/10.2136/sssaj2004.1501Coppola A., Dragonetti G., Comegna A., Lamaddalena N., Caushi B., Haikal M.A., Basile A. (2013): Measuring and modeling water content in stony soils. Soil & Tillage Research, 128: 9–22.CSI (1996): CS615 Water Content Reflectometer: Service Manual. Logan, Campbell Scientific, Inc.Fan Junliang, Scheuermann Alexander, Guyot Adrien, Baumgartl Thomas, Lockington David A. (2015): Quantifying spatiotemporal dynamics of root-zone soil water in a mixed forest on subtropical coastal sand dune using surface ERT and spatial TDR. Journal of Hydrology, 523, 475-488 https://doi.org/10.1016/j.jhydrol.2015.01.064Francesca Ventura, Osvaldo Facini, Stefano Piana, Paola Rossi Pisa (2010): Soil Moisture Measurements: Comparison of Instrumentation Performances. Journal of Irrigation and Drainage Engineering, 136, 81-89 https://doi.org/10.1061/(ASCE)0733-9437(2010)136:2(81)Jones Scott B., Wraith Jon M., Or Dani (2002): Time domain reflectometry measurement principles and applications. Hydrological Processes, 16, 141-153 https://doi.org/10.1002/hyp.513Kelleners T. J., Seyfried M. S., Blonquist J. M., Bilskie J., Chandler D. G. (2005): Improved Interpretation of Water Content Reflectometer Measurements in Soils. Soil Science Society of America Journal, 69, 1684- https://doi.org/10.2136/sssaj2005.0023Kim K.C., Benson C. (2002): Water Content Reflectometer Calibration for Final Cover Soils. Geo-Engineering Report No 02-12, Madison, University of Wisconsin-Madison.Kim K.C., Hwang K.H., Lee S. (2005): A study on the measurement of volumetric water content using WCR in final cover layer of landfill. Journal of the Korean Geotechnical Society, 21: 147–154. (in Korean)Liu Q., Shalaby A. (2013): Volumetric moisture content analysis and the implementation of spring load restriction of low volume road. In: Al-Qadi I.L., Murrell S. (eds): Conf. Proc. Airfield and Highway Pavement 2013, Los Angeles, June 9–12, 2013: 1316–1327.Munoz-Carpena R. (2012): Field Devices for Monitoring Soil Water Content. Agricultural and Biological Engineering Department, University of Florida, BUL343. Available at http://edis.ifas.eduOr Dani, Wraith Jon M. (1999): Temperature effects on soil bulk dielectric permittivity measured by time domain reflectometry: A physical model. Water Resources Research, 35, 371-383 https://doi.org/10.1029/1998WR900008Orton T.L. (2002): Development of a test cell to evaluate embankment infiltration. In: WM’02 Conference, Tucson, Feb 24–28, 2002.Quinones H., Ruelle P., Nemeth I. (2003): Comparison of three calibration procedures for TDR soil moisture sensors. Irrigation and Drainage, 52, 203-217 https://doi.org/10.1002/ird.95Reder Alfredo, Rianna Guido, Pagano Luca (2014): Calibration of TDRs and Heat Dissipation Probes in Pyroclastic Soils. Procedia Earth and Planetary Science, 9, 171-179 https://doi.org/10.1016/j.proeps.2014.06.016Seyfried M.S., Murdock M.D. (2001): Response of a New Soil Water Sensor to Variable Soil, Water Content, and Temperature. Soil Science Society of America Journal, 65, 28- https://doi.org/10.2136/sssaj2001.65128xSeyfried Mark S., Murdock Mark D. (2004): Measurement of Soil Water Content with a 50-MHz Soil Dielectric Sensor. Soil Science Society of America Journal, 68, 394- https://doi.org/10.2136/sssaj2004.3940Skierucha Wojciech, Wilczek Andrzej, Alokhina Olga (2008): Calibration of a TDR probe for low soil water content measurements. Sensors and Actuators A: Physical, 147, 544-552 https://doi.org/10.1016/j.sna.2008.06.015Stander Emilie K., Rowe Amy A., Borst Michael, O’Connor Thomas P. (2013): Novel Use of Time Domain Reflectometry in Infiltration-Based Low Impact Development Practices. Journal of Irrigation and Drainage Engineering, 139, 625-634 https://doi.org/10.1061/(ASCE)IR.1943-4774.0000595Stangl R., Buchan G.D., Loiskandl W. (2009): Field use and calibration of a TDR-based probe for monitoring water content in a high-clay landslide soil in Austria. Geoderma, 150, 23-31 https://doi.org/10.1016/j.geoderma.2009.01.002Stenger Roland, Barkle Greg, Burgess Craig (2005): Laboratory calibrations of water content reflectometers and their in-situ verification. Australian Journal of Soil Research, 43, 607- https://doi.org/10.1071/SR04177Sotelo M., Mazari M., Garibay J., Nazarian S. (2014): Variability of moisture content measurement devices on subgrade soils. In: Proc. Geo-Congress 2014, Atlanta, Feb 23–26, 2014: Technical Paper, GSP 234: 1425–1432.Topp G. C., Yanuka M., Zebchuk W. D., Zegelin S. (1988): Determination of electrical conductivity using time domain reflectometry: Soil and water experiments in coaxial lines. Water Resources Research, 24, 945-952 https://doi.org/10.1029/WR024i007p00945Udawatta Ranjith P., Anderson Stephen H., Motavalli Peter P., Garrett Harold E. (2011): Calibration of a water content reflectometer and soil water dynamics for an agroforestry practice. Agroforestry Systems, 82, 61-75 https://doi.org/10.1007/s10457-010-9362-3Visconti Fernando, de Paz José Miguel, Martínez Delfina, Molina Mª José (2014): Laboratory and field assessment of the capacitance sensors Decagon 10HS and 5TE for estimating the water content of irrigated soils. Agricultural Water Management, 132, 111-119 https://doi.org/10.1016/j.agwat.2013.10.005Walker Jeffrey P, Willgoose Garry R, Kalma Jetse D (2004): In situ measurement of soil moisture: a comparison of techniques. Journal of Hydrology, 293, 85-99 https://doi.org/10.1016/j.jhydrol.2004.01.008Western Andrew W., Seyfried Mark S. (2005): A calibration and temperature correction procedure for the water-content reflectometer. Hydrological Processes, 19, 3785-3793 https://doi.org/10.1002/hyp.6069Zegelin S. J., White I., Jenkins D. R. (1989): Improved field probes for soil water content and electrical conductivity measurement using time domain reflectometry. Water Resources Research, 25, 2367-2376 https://doi.org/10.1029/WR025i011p02367